Voltage and Current sensing from evive menu
evive can be used as a voltmeter or an ammeter. It uses 24-bit dual channel ADC IC (ADE7912) for sensing Analog signals . There are two channels for data acquisition namely Probe I/V and Probe V. The Probe I/V can be used to sense currents or voltage as per the Jumper selection(JP 2 on evive). The current sensing has accuracy of 3mA with range of -3A to +3A, while in voltage mode, the accuracy is of 3mV with range of -5V to +5V. The Probe V can be used for higher voltage sensing in range of -30V to +30V with accuracy of 10mV.
There are three header and three banana jacks where you can connect your sensing wire and sensing banana cable respectively . The middle one is the reference point (means the voltage or the current is measured with reference to this probe) in both cases for headers as well as banana jacks.The banana jacks are named Probe I/V, COM and Probe V respectively. Actually both header and banana jacks are same connections. Just to provide user with an ease of giving input signal through jumper wires or commonly used banana cables two different locations for applying signal are given. The header pins are marked as "Graph" on evive board. In that also pin on left side is Probe I/V, middle one is COM and on right side is Probe V.
There are two options for user depending upon location jumper JP2 on evive:
- ProbeV (30V) and I/V (5V): When jumper JP 2 is kept on V side(right side) at that time we can measure a voltage signal in range of -5V to +5V on Probe I/V with accuracy of 3mV and a voltage signal in range of -30V to +30V on ProbeV.
- ProbeV (30V) and I/V (3A): When jumper JP 2 is kept on I side(left side) at that time we can measure a current signal in range of -3A to +3A on Probe I/V with accuracy of 3mV and a voltage signal in range of -30V to +30V on ProbeV .
You have to be very careful about the jumper for sensing. If you are sensing voltage from ProbeI/V channel, then it should be placed on right header(V side) and if you sensing current, then place it on the left header(I side).